ABBOTT CELL DYN SAPPHIRE PDF
Korean J Lab Med. Jun;27(3) [Evaluation of the Abbott Cell-Dyn Sapphire hematology analyzer]. [Article in Korean]. Park Y(1), Song J, Song S. Name of instrument, CELL-DYN Sapphire*. First year installed in U.S./Outside U.S./No. of units sold in , //—. No. units installed in U.S./Outside. The CELL-DYN Sapphire hematology system is designed to address routine testing needs of high-volume hematology laboratories and difficult.
|Published (Last):||27 May 2007|
|PDF File Size:||7.55 Mb|
|ePub File Size:||20.53 Mb|
|Price:||Free* [*Free Regsitration Required]|
Comparative evaluation of the Iris iQ body fluid module with manual hemacytometer count. BMononuclear cells with the cytocentrifuged preparation possibly including epithelial cells and macrophages.
Prices and Information shown are for reference only and may change without notice. For CSF specimens, many cases were falsely classified as abnormal, leading to a very poor specificity of The counting and differentiation principles are based on an optical and an impedance channel.
MAPSS plus 3 color fluorescent technology. Related articles in Web of Science Google Scholar. RBC and WBC counts were compared with the conventional manual hemacytometer counts, whereas leukocyte differential counts were compared with microscopic counts of cytocentrifuged preparations.
More about Abbott Diagnostics. Inquire Here Reagent Type: Inquire Here Sample Size Max: Inquire Here Reagent System: Save time Submit your details once and make multiple inquiries. Manuals and Resources Note: In the present article, we describe its applicability for the distinct body fluids, and we highlight limitations and caveats. However, manual hemacytometer counts could even be more imprecise at low cell concentrations.
Different cell concentrations were prepared, and 10 aliquots of each cell concentration were counted. Citing articles via Web of Science They include programmable automated alarm systems for indicating results outside the reference range. The poor reproducibility at low cell concentrations that we encountered is most likely to be held responsible for the absence of agreement with the manual microscopic count.
All samples were collected into sterile tubes without anticoagulants. Inquire Here Reagent Delivery: Simple, automated, single mode calibration makes calibrating the analyzer a quick and easy operation.
The Cell-Dyn Sapphire is an automated hematology analyzer designed for counting peripheral blood cells in whole blood samples.
Write your own review. Apart from the organizational and inherent economic disadvantages, the manual method also has poor precision and large interobserver variability.
It furthers the University’s objective of excellence in research, scholarship, and education by publishing worldwide. Likewise, a very good negative likelihood ratio 0.
For current regulatory status on products within this website, please contact your sales representative. Diamond Diagnostics expressly disclaims any affiliation with products it does not manufacture, as well as sponsorship by other manufacturers.
[Evaluation of the Abbott Cell-Dyn Sapphire hematology analyzer].
Full data management with data storage for 10, cycles with graphics for easy review of patient results. Similarly, an excellent positive likelihood ratio Assays Total assays tested: Use of the Advia hematology analyzer in differential cytologic analysis of biological fluids cerebrospinal, peritoneal, pericardial, synovial, and others.
This helps other members to better understand the Reviewer’s experience and expertise. Each dilution was examined 10 times.
Email alerts New issue alert. We evaluated the analytic and clinical performance of the Cell-Dyn Sapphire for blood cell counting and WBC differential count in body fluids.
Abbott Diagnostics CELL-DYN SapphireÂ® | Forums | Questions | Discussions | Help | LabWrench
If you find a document or resource that does not belong please let us know by emailing info labwrench. Experienced observers examined at least nucleated cells.
Wilcoxon tests for paired samples were used because normal distribution could not be demonstrated.
Abbott products have helped transform the practice of medical Cell Dyn Number of Assays: September 25 th1 Reply Post a Reply. It is a multi-parameter automated hematology analyzer designed for in vitro diagnostic use in counting and characterizing blood cells.
Cell Dyn Sapphire – Abbott – Hematology Analyzer
International Council for Standardization in Haematology: Furthermore, we do not recommend automated differentiation of NCs by the Cell-Dyn Sapphire because of its variable analytic performance in the presence of epithelial cells and macrophages for the distinct body fluids. When the results were completely determined following standard operating procedures, the RBC count, WBC count, and WBC differential count were performed on the leftover samples, provided there was a sufficient amount of residual sample.
Agreement between the automated cell counts and the microscopic data was examined by Passing-Bablok regression analysis. We appreciate the support of individuals and companies making their documents and resources available to our community.
The names and logos of manufacturers, their instruments, and their products referred to herein may be protected by trademark or other law, and are used herein solely for purpose of reference.